AQMesh co-location comparison trial results for CO in Belgium, Europe (V4.2.3)
|Correlation co-efficient (R2)
|December 2017 - January 2018
|AQMesh processing algorithm
|Averaged sample interval
All comparison trials are from regularly maintained reference stations. Not all are specifically identified due to the administrative overhead of getting permission to name data sources. We do not always use ratified reference data to avoid delays in analysis but re-running the comparison analysis after data ratification often improves correlation.
AQMesh co-location comparison trial results for CO in Southern USA (V4.2.3)
AQMesh co-location comparison trial results for NO in California, USA (V5.0)
AQMesh co-location comparison trial results for NO in London, UK (V4.2.3)
AQMesh co-location comparison trial results for NO2 in California, USA (V5.0)
AQMesh co-location comparison trial results for NO2 in London, UK (V4.2.3)
AQMesh has been independently tested and verified in numerous academic studies and publications
A wide range of private, academic and public organisations across the globe are currently using AQMesh
AQMesh air quality monitoring systems can measure just one gas or up to six gases, particles, noise and wind speed & direction. Humidity, pressure and pod temperature are always measured.
The AQMesh air quality monitor is neat, compact and easy install within just a few minutes and experienced technical staff are not required.
With wireless communications using the global mobile network and a choice of power options, including batteries, mains supply or a bespoke solar pack, AQMesh air quality monitors can be installed in almost any location.
With AQMesh outdoor air quality monitors, meaningful correction of cross-gas effects and interference from environmental conditions is consistently delivered through proven and traceable processing.